Amateur radio, RF design, electronics, uC, software, hacks, technology
I have posted the equivalent schematic for my hipot tester.
The unit consists of a 700VA 12KV AC RMS output (nominal) center tapped neon sign transformer that I purchased on Ebay. The trafo was designed for less than 230V AC RMS nominal primary voltage. Therefore the transformer gives out a bit more than 12KV AC RMS when supplied from the 230V AC RMS mains in Norway. After the rectifier I get approx 17-18 KV DC. The transformer is simulated by a sine voltage source in LTSpice. The transformer is insulated from ground in the tester for safety reasons. The schematic shows a gnd connection but that is only to get the Spice 3F4 solver to accept the netlist. It is recommended to insulate your tester from ground if you build one. There is a variac (not drawn on the schematic) in front of the transformer and the input is current limited by two 100W light bulbs in parallel (bulbs not drawn on the schematic). The voltage metering circuit is a uA meter with a shunt resistor and the current measurement instrument is a std. 100uA instrument protected by back to back diodes (not drawn on the schematic). The instruments are at high potential so adequate protection must be implemented to prevent arcs to the operator. The device under test (DUT) is drawn. Note that a working tube has much higher resistance than 100 megs that is the load in the simulation. However this is placed as a load test for the simulation to check smoothing of the output voltage. The chassis is built in plastic and lexan and I have ensured long creepage distances by utilisation of vertical lexan insulators. There are no detected creepage currents at all even at full DC HV potential with my current design.
WARNING: do not attempt to build this high voltage device unless you are experienced in electronics and high voltage circuits. Touching the HV leads may instantly kill you. I do not accept any responsibility for your use of this device. The schematic is not a complete circuit description.
Download the LTspice simulation file here: LTspice schematic of hipot tester
Video of the hipot tester in operation (test of an air variable capacitor):